Statistics from Altmetric.com
The recent article by Pisetsky et al1 showing data derived from a comparison of different antinuclear antibody (ANA) assays in a cohort of patients with established systemic lupus erythematosus triggered significant discussions among experts in ANA testing. In this context, we read with interest the correspondence by van Hoovels et al2 describing variation in ANA detection by automated indirect immunofluorescence (IIF) analysis. The data presented showed significant variation between manufacturers and also between laboratories and even within the same laboratory, which raised concerns. The majority of laboratories (n=12) used NOVA View (Inova Diagnostics), followed by G-Sight (Menarini, n=9; split into HEp-2 and HEp-2000) and Image Navigator (ImmunoConcepts, n=2).
A similar study has been conducted as part of the regulatory clearance of the NOVA View system, which showed significantly higher consistency. In addition, van Hoovels et al2 had previously performed a similar study that also reported variability higher than expected.3 However, in this study, the variability could be improved by …
If you wish to reuse any or all of this article please use the link below which will take you to the Copyright Clearance Center’s RightsLink service. You will be able to get a quick price and instant permission to reuse the content in many different ways.